NOTES: Next day delivery on the PT900 is available to mainland UK when rental paperwork is received by 12 midday (stock permitting).
Whatever your experience with ultrasonic flow meters, we are confident that our quick start guides and technical support engineers should be able to get you up and running in minutes. The new and improved user interface makes it easier than ever to program the meter for your application. The app even has a range of instruction videos should you need any extra guidance. These videos can also be viewed in the FAQs section below.
Manufactured by: GE Panametrics
Model: Rental PT900 Flow Meter
Cooling and heating water
|Flow Range||-40 to 40 ft/s (-12.2 to 12.2 m/s) biderectional|
Pipe ID>6 in (150 mm): ±1% to 2% of reading
Pipe ID<6 in (150 mm): ±2% to 5% of reading
|Repeatability||±0.1% to 0.3% of reading|
0.5 in to 300 in (12.7 mm to 7.6 m) and large
|Pipe Wall Thickness||Up to 3 in (76.2 mm)|
All metals and most plastics. Consult GE for concrete, composite materials, and highly corroded or lined pipes.
|Clamp-On Flow Accuracy (Velocity)||
Pipe ID>6 in (150 mm): ±1% to 2% of reading typical
|Datalogging||Memory capacity to log over 100,000 flow data point|
|Output||- One 0/4 to 20 mA current output, 550 W
- One user-selectable pulse (solid state, 5 V
maximum) or frequency (5 V square wave, 100 to 10,000 Hz)
- Two 4 to 20 mA analog inputs with switchable
power supply for loop-powered temperature
|Power||Built-in battery with external 12v input, 9 - 11 hours operation|
|Battery Charger Input||100 to 250 VAC, 50/60 Hz, 0.38|
|Dimensions||(h x w x d) 9.4 in x 5.5 in x 1.5 in
(238 mm x 138 mm x 38 mm)
3 lb (1.36 kg)
How do I pair the PT900 with the Android Tablet?
How do I setup the flow meter and download the logs?
How do I load the PT900 App from SD card?
How do I load the PT900 App from Google Play?
How do I install the transducers and clamping fixtures?